Cable interface between laptop and Philips MX800 #292
Gavriel Jacoby (2020-07-27 10:35) · 14 view(s) · 1 comment(s)
Is it possible to connect a Philips patient monitor MX800 using its RS232 ethernet cable directly to the network socket on a laptop?
This did work using the ixtrend free download sample.
vitaldb2017 (2020-07-28 21:18)
Vital Recorder supports serial communication. You need to connect to the laptop using a serial port or serial to USB converter.